[an error occurred while processing this directive]
Posted by mike on February 25, 2002 at 07:18:00:
In Reply to: Re: calorimeter and sterilizers posted by mike o'malley on February 25, 2002 at 07:13:24:
I am having the same trouble. I can fax you a steam quaility test procedure from a Amsco customer training manual if you want. It details all the procedures that are needed to do a quality of steam test. : : i'm having a problem with wet loads in our sterilizers at the hospital where i work. the Castle company says the machines spec out! I have repiped the steam feeds to the machines with 1} spirax separator with f&t traps
: : 2} new spence prv's, and steam filters with thermo-dynamic traps, machines continue to do wet loads, castle co maintains we have wet steam, I need to test the staem at the sterilizers and am told the way to do this is by a calorimeter! I can find no one in my area who has any or much knowledge of this, this has become a key issue at the hospital and any help would be greatly appreceted !!
Post a Followup